You can use grid scans to automatically centre your sample. This method is very successful with well frozen (ice-free) crystals mounted in clean loops.
For an already mounted sample make sure your loop is in the field of view and then click the X-ray Centre button on the Align tab of the Alignment view in GDA.
To enable X-ray Centring when driving data collections for each robot loaded sample, please open the 'User Options' and click the option labelled 'HT X-Ray Centring (5 min/sample)' before you load the (first) sample you want to centre. Additionally, the second option 'Use OAV Centring after Keyence Centring' needs to be ticked, which is the default.
Before selecting a sample make sure you specify your data collection parameters in the data collection table, as the data collection currently starts immediately after X-ray Centring.
If you now load your sample, the routine puts the loop at the cross-hairs (Keyence Centring), performs the OAV centring and then automatically carries out a two-dimensional grid scan. Afterwards, a finer, linear grid scan is performed perpendicularly to the two-dimensional one at the position where the crystal diffracts best. Then, the data collection is automatically started. The whole process from loading the sample to the end of a 360 degree data collection with standard parameters takes approximately 3.5 min.
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2018 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus