Sample centring is now managed per sample, and can be edited in the Data collection table.
The Sample Alignment perspective has been removed as part of these changes and the centring options moved to the OAV tab in all perspectives where appropriate.
When manually loading a sample from the sample perspective, the sample will be loaded then loop centred. No further action will be performed unless requested.
When loading samples from the data collection perspective, whether from the data collection form/table or the screening tab, you now have a dropdown box that lists the type of centring to be used prior to data collection.
This dropdown gives the choice of several centring options:
If you have already centred your sample then choose None to simply proceed with a single data collection/screening.
For queue mode you must select one of the Auto options, we recommend Auto X-ray which will take approximately [2 minutes + total exposure time] per sample. E.g. a queue with a 7s exposure time would take around 2min 7s in total for each sample (as measured from one robot load to next robot load).
Auto loop is only suitable for cases where the crystal size and shape match the loop.
You must make sure the appropriate sample centring mode is selected as part of your data collection form. The choice is literal and GDA will do exactly what you ask for.
If you have spent a lot of time and effort manually centring your sample then please double check that None is selected prior to collecting data. Otherwise your crystal will move.
Be careful when switching between screening and data collection tabs as the settings on the form are independent on each page.
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2020 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus
Diamond Light Source® and the Diamond logo are registered trademarks of Diamond Light Source Ltd