Beamline Phone Number:
+44 (0) 1235 778707
8th June 2020: Access to MX beamlines is currently operating under new guidelines. Please read carefully.
I03 is one of five macromolecular crystallography (MX) beamlines at Diamond. I03 is a tuneable beamline with a working wavelength range of 0.6 - 2.48 Å. The standard working wavelength is 0.976 Å (12.7 keV) with a focused beam size of 90 x 30 um (FWHM). As with all the MX beamlines, I03 uses SPINE standard pins and Unipucks for cryo-cooled samples.
I03 can provide containment measures for experiments involving biological agents in Hazard Groups 2 and 3.
Please note there are some adjustments required for Data Processing from i03 Eiger Data from mid 2021.
|Standard energy (Å / keV)||0.98 / 12.7|
|Flux at 12.7 keV, focus at sample (ph/s)||1.7 x 1012|
Beam size options:
90 x 20 µm
90 x 30 µm
100 x 100 µm
55 x 10 µm
|Wavelength range (Å)||0.6 - 2.3|
|Energy range(keV)||5.2- 21.0|
|Detector||Eiger2 XE 16M|
|Time to collect 180 ° data set (seconds)||3.2 +|
|No. of pins in dewar||592|
|Sample exchange time (seconds)||18|
|Guide to samples screened/hour (X-ray Centring)||15|
+ 1800 frames, 0.1 ° oscillation at 560Hz
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2020 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus
Diamond Light Source® and the Diamond logo are registered trademarks of Diamond Light Source Ltd
Registered in England and Wales at Diamond House, Harwell Science and Innovation Campus, Didcot, Oxfordshire, OX11 0DE, United Kingdom. Company number: 4375679. VAT number: 287 461 957. Economic Operators Registration and Identification (EORI) number: GB287461957003.