Angle-Resolved Photoemission Spectroscopy (ARPES) maps the dispersion of electronic bands near the Fermi level and, in particular, the Fermi surface itself by exciting the bound electrons in a metal with a given photon energy hv. The momentum parallel to the surface is fully conserved, thus making the method suitable for layered low-dimensional materials. The three-dimensional momentum distribution is also reflected in the photoelectron features thus making the spectroscopy applicable to metallic single crystals, provided that a well-defined clean surface can be prepared in ultra-high vacuum. The minimum samples size is 500 x 500 μm2 given by the light spot (50 x 50 μm2) and the sphere of confusion of the sample goniometer.
The fine structure of the energy spectra of photoelectrons reflect the interaction of the photo-hole left behind with the re-arranging electronic and vibronic structure in the sample. The width of the band as a function of binding energy corresponds to the imaginary, the deviation from a bare dispersion to the real part of the spectral function. High sensitivity to excitations on a thermal energy scale require measurements at low temperatures (less than 10 K) at an energy resolution of a few meV (2 – 5 meV) and a momentum resolution at a fraction of the Brillouin zone dimensions (0.1° angular resolution).
Spectro-microscopy combines the power of momentum and energy resolution in the electron analyser with a microscopic light spot and raster sample scanner. The reduced transmission of the nano-focusing optics necessitates the reduction of energy resolution to about 30-50 meV, but the power of electron band mapping allows either a highly selective contrast in microscopy or a band and Fermi surface mapping at high spatial resolution. It can thus serve to image grains, domains of artificial structure of micro- or nanometric dimensions or to illuminate selectively micro- and nano-domains for ARPES analysis.
At the same time the instrument is used a lot as an ARPES spectrometer with sub-micron spot.
Samples have to be prepared in situ in ultrahigh vacuum to achieve atomically controlled, clean surfaces. The following methods will be provided in the vacuum system of HR-ARPES or nanoARPES:
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2020 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus
Diamond Light Source® and the Diamond logo are registered trademarks of Diamond Light Source Ltd
Registered in England and Wales at Diamond House, Harwell Science and Innovation Campus, Didcot, Oxfordshire, OX11 0DE, United Kingdom. Company number: 4375679. VAT number: 287 461 957. Economic Operators Registration and Identification (EORI) number: GB287461957003.