I03 is one of five macromolecular crystallography (MX) beamlines at Diamond. I03 is a tuneable beamline with a working wavelength range of 0.6 - 2.48 Å. The standard working wavelength is 0.976 Å (12.7 keV) with a focused beam size of 90 x 30 um (FWHM). As with all the MX beamlines, I03 uses SPINE standard pins and Unipucks for cryo-cooled samples.
I03 can provide containment measures for experiments involving biological agents in Hazard Groups 2 and 3.
I03 and I04 are equipped with Eiger2 XE 16M detectors - collect at up to 560 Hz!
New centring options, choose from 'Manual', 'Loop' or 'X-ray'. Info, scroll to end of page.
Beamline workstations have been upgraded to RedHat 7, more info here
Latest generation Eiger2 X 16M detector on I03. More information here.
The SmarGon is back from repair, but still has "issues". The chi angle can be changed, but the crystal will need to be re-centred after changing it. The phi rotation is currently unavailable.
New robot dewar installed - now holds 37 puck, 592 samples.
We will be barcoding pucks from April 2017 onwards.
Data reprocessing now available in ISPyB. See here
New remote access connection available
Go to GDA - Messages- Beamline notes to read the message of the day. Click here for more.
Loan pucks available upon request, check here
Test your network connection to Diamond with our SpeedTest
Remote access available on all MX beamlines at all times since January 2013
|Standard energy (Å / keV)||0.98 / 12.7|
|Flux at 12.7 keV, focus at sample (ph/s)||1.7 x 1012|
Beam size options:
90 x 20 µm
90 x 30 µm
100 x 100 µm
55 x 10 µm
|Wavelength range (Å)||0.6 - 2.3|
|Energy range(keV)||5.2- 21.0|
|Detector||Eiger2 XE 16M|
|Time to collect 180 ° data set (seconds)||3.2 +|
|No. of pins in dewar||592|
|Sample exchange time (seconds)||18|
|Guide to samples screened/hour (X-ray Centring) ||15|
+ 1800 frames, 0.1 ° oscillation at 560Hz
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2018 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus