RCaH equipment can also be used by Diamond users and Diamond Staff, for more information please visit:
This overpressure glove box is maintained for the transfer of dry materials, such as alkali metals under very low oxygen, water and solvent levels. The use of non-aqueous solvents is allowed. It also includes a balance and pellet press. Gas used is ambient Argon.
- Programmable logic control touch screen system
- Large round anti-chamber with sliding tray & round mini antechamber.
- Regenerable purifier unit. Gas –Ar/H2 mixture (H2 3-5%).
- Attainable purity: - H2O <0.1ppm; O2 <0.1ppm; activated carbon solvent scrubber removes solvent vapours but levels are not measured.
Glovebox Balance: 3 decimal place balance for weighing mg - g samples
Heat sealer available upon request
Small volumes of solvent are permitted
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2022 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus
Diamond Light Source® and the Diamond logo are registered trademarks of Diamond Light Source Ltd
Registered in England and Wales at Diamond House, Harwell Science and Innovation Campus, Didcot, Oxfordshire, OX11 0DE, United Kingdom. Company number: 4375679. VAT number: 287 461 957. Economic Operators Registration and Identification (EORI) number: GB287461957003.