I03 is one of five macromolecular crystallography (MX) beamlines at Diamond. I03 is a tuneable beamline with a working wavelength range of 0.6 - 2.48 Å. The standard working wavelength is 0.976 Å (12.7 keV) with a focused beam size of 80 x 20 um (FWHM). As with all the MX beamlines, I03 uses SPINE standard pins and Unipucks for cryo-cooled samples. I03 also offers in situ experiments using SBS format plates.
I03 can provide containment measures for experiments involving biological agents in Hazard Groups 2 and 3.
BART upgrade - can now store 30 pucks in the dewar
New remote access connection available
Go to GDA - Messages- Beamline notes to read the message of the day. Click here for more.
New Pilatus3 6M detector installed in April 2014. Minimum exposure time 0.01s.
Loan pucks available upon request, check here
Test your network connection to Diamond with our SpeedTest
Updated excel spreadsheet for shipments in ISPyB. Download and use the latest version.
Remote access available on all MX beamlines at all times since January 2013
|Standard energy (Å / keV)||0.98 / 12.7|
|Flux at 12.7 keV, focus at sample (ph/s)||1.7 x 1012|
Beam size options:
90 x 20 µm
90 x 30 µm
100 x 100 µm
55 x 10 µm
|Wavelength range (Å)||0.6 - 2.3|
|Energy range(keV)||5.2- 21.0|
|Time to collect 180 ° data set (seconds)||18 +|
|No. of pins in dewar||480|
|Sample exchange time (seconds)||18|
|Guide to samples screened/hour (user centering)||30|
+ 1800 frames, 0.1 ° oscillation at 100 Hz
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2016 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus