Beamline Phone Number:
+44 (0) 1235 567473
Principal Beamline Scientist:
Tel: +44 (0) 1235 778156
Tel: +44 (0) 1235 778056
The high field magnet system is an experimental station providing very high magnetic field and low temperature environment optimized for studies of magnetism using x-ray dichroism.
The magnetic field is directed along the direction of the x-ray beam. The system can be fitted with one of two sample helium cryostats (inserts). Each cryostat can vary the sample temperature within a specific temperature range.
Main system specifications are summarized in the following table:
|Maximal magnetic flux density||± 14 T|
|4He insert||3 - 420 K|
|3He insert||not currently available|
|Field polarity reversal time||< 1 hour|
|Sample rotation around the vertical axis||± 90°|
The sample will be placed in ultra high vacuum.
Dichroism effect in absorption will be detected by total electron yield and fluorescence yield.
The total electron yield is more surface sensitive. This is achieved by grounding the electrically isolated sample through a sensitive current amplifier.
The fluorescence yield gives more sensitivity to the bulk of the sample. This is achieved with a photodiode placed close to the sample.
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2022 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus
Diamond Light Source® and the Diamond logo are registered trademarks of Diamond Light Source Ltd
Registered in England and Wales at Diamond House, Harwell Science and Innovation Campus, Didcot, Oxfordshire, OX11 0DE, United Kingdom. Company number: 4375679. VAT number: 287 461 957. Economic Operators Registration and Identification (EORI) number: GB287461957003.