Small Angle Scattering for Engineering and Material Science is mostly done with neutrons, because the available SAXS instruments do not offer the required penetration lengths. I12 will operate at 50-150 keV, offering potential for SAXS at high photon energies that so far has not been exploited. SAXS measurements of correlation on the nanometre-scale could reveal microstructural information that is essential to Engineering and Material Science, particularly for the characterization of deformation on the grain and sub-grain level. The technique is complementary to imaging as it offers a higher resolution in real-space units.
The key-elements for high energy x-ray small angle scattering are now available: efficient detectors with good spatial resolution (10-30 µm) at high energies, focusing optics and a high brilliance source for the photon energy range between 50 keV and 150 keV.
In the first experimental hutch of I12, it will be possible to place a 2D detector at a distance of 4.5 m from the sample. Larger sample-to-detector distances of around 30 m could be set up with the detector in the third optics hutch, located between the first and the second experimental hutch.
Rat tail collagen SAXS standard
Fig: Pattern from rat tail collagen SAXS standard, showing diffraction orders 1 to 11. Image taken with large area 2D detector during commissioning experiment, beam energy approx. 80 keV.
Diamond Light Source is the UK's national synchrotron science facility, located at the Harwell Science and Innovation Campus in Oxfordshire.
Copyright © 2018 Diamond Light Source
Diamond Light Source Ltd
Harwell Science & Innovation Campus